- for manual or semiautomated disinfection
- processing of pre-cleaned flexible endoscope
- ophtalmological equipment
- surgical instruments made of stainless steel
- heat sensitive medical devices
gigasept® PAA is a ready to use high-level disinfectant suitable for disinfection of a variety of medical devices and instruments
Incompatible materials: iron, toolsteel, aluminium, platinum, nickel-/ chromeplated or non ferrous metal alloys parts (e.g. copper and brass)
Compatible materials: (limited to the maximum contact time; obligatory rinse after contact) polycarbonate, polyurethane, silicone, polysulfone, polyamide, polyester, polystyrene
gigasept® PAA is ready to use, active immediately and stable over 7 days or 50 cycles. The solution can be tested using gigasept® PAA test stripes.
Always ensure the pre-cleaning of the endoscope and a complete wetting of the surfaces. Immerse hollow objects in such a way that air escapes completely from the lumen.
After the necessary contact time rinse carefully with sterile or filtered water before re-use.
Do not exceed the recommended contact time.
Avoid the mixing of gigasept® PAA with other products (cleaner, disinfectants), particularly with aldehydes or derivates of halogens (e.g. glutaral, sodium hypochlorite, etc.)
|Microorganism||EN standards||Contact time|
|Bacteria||EN 1040, EN 13727 (clean conditions)||5 mins|
|Mycobacterium terrae||EN 14348||5 mins|
|Fungi||EN 1275, EN 13624||5 mins|
|Spores||EN 14347 (clean conditions)||5 mins|
|Viruses||EN 14476||5 mins|
100g of gigasept® PAA contains: 0,1 g Peracetic acid
Other ingredients: corrosion inhibitors, surfactants, stabilising agents, excipients
R41: Risk of serious damage to eyes.
S26: In case of contact with eyes, rinse immediately with plenty of water and seek medical advice.
S39: Wear eye/face protection.
USE DISINFECTANT SAFELY. ALWAYS READ THE LABEL AND USE ONLY AS DIRECTED.
For external use only. Do not swallow.
Appropriate personal protection equipment are required
Wearing of gloves are recommended
Avoid contact with eyes. If in eyes, rinse well with water.
Do not mix with detergents or other chemicals.
Do not use after the expiry date.
|Item||Delivery form||Item no.|
|gigasept® PAA 5 L||1/Carton||195 751|
|gigasept® PAA test strips (50 pcs)||195 704|
|Dosing tap (5 L canister)||135 501|