gigazyme® is an enzymatic cleaning agent designed specifically for the manual and semi-automatic cleaning of endoscopes and for the manual cleaning of surgical instruments.The ideal combination of high-quality enzymes (Protease = cutting proteins, Lipase = cutting fat, Amylase = cutting starch) and an innovative system of sufactants allows excellent cleaning of thermostable and thermolabile instruments of any kind. gigazyme® is also ideal for use in an ultrasonic bath.
gigazyme® is a triple-enzyme cleaner designed specifically for the manual and semi-automatic cleaning of endoscopes and for the manual cleaning of surgical instruments.
gigazyme® can be used in immersion baths and ultrasonic baths.
gigazyme® is a concentrated product and is diluted with water to the desired working concentration. Keep the water temperature below 45°C to avoid protein denaturation.
The dosing quantity depends on how the instruments are reprocessed and the level of soiling. The following parameters are recommended when using gigazyme®:
Normal contamination: 0.5 - 2% (1 litre of a 1% solution corresponds to 990 ml water and 10 ml gigazyme®. Severe contamination: up to 5%.
Immerse instruments to be reprocessed into the working solution. Ensure items are completely wetted - also in the case of hollow instruments - and allow them to soak in the solution until the instruments are clean on visual inspection After cleaning, thoroughly rinse/flush the medical devices with water that has at least drinking water quality to thoroughly remove any residues of the solution.
Please note the treatment recommendations provided by the instrument manufacturers.
We recommend replacing the working solution every working day at least, or immediately if there is visible soiling.
5-15% non-ionic surfactants, enzymes, fragrances.
Always read the label and product information sheet before use.
This product is intended for use by professionals only.
Any serious incidents associated with the product should be reported to the manufacturer and competent authority.
For optimal results, the temperature of the cleaning solution should be kept below 35°C. Failure to do so may cause blood to coagulate, particularly when instruments and endoscopes contaminated with blood are cleaned in an ultrasonic bath. The solution becomes turbid when it reaches approximately 32°C, indicating that the water temperature is too high. This turbidity does not have a negative impact on cleaning performance.
Do not mix gigazyme® with other products.
|Item||Delivery form||Item no.|
|gigazyme -int- 2 l FL||5/Carton||on request|
These products are not available in every country. For more information please contact our local subsidiary or distributor.
schülke manufactures products economically and with advanced, safe and environmentally friendly production processes while at the same time maintaining our high quality standards.